However, there are some specifications to follow such as the size of the patches.
We recommend a ration of 2
Remember to select the correct reading modes
The measurement of the range being done patch by patch, it would be a shame to notice at the end of the measurement, that you were mistaken in the choice of the SCI / SCE or UV100 / UV CUT400 mode especially after having measured 1000 patches. ..
8) Measurement of samples in a "Color Table"
The method for adding samples to a color table is the same as for other spectrophotometers.
Mode UV CUT400: Measurement with UV filtered below 400 nm to ignore optical brighteners
Example of measurement carried out on a shiny Chromaluxe plate
A Gray sample was measured with the four reading combinations: The display of the reflectance curves allows you to view the difference in measurement between the different modes:
- SCI - UV100 (Red)
- SCE - UV100 (Blue)
- SCI - UV CUT400 (Orange)
- SCE - UV CUT400 (Cyan)
Note: On the right column you can directly obtain the information associated with the selected sample.